We know how challenging multi-color Immunofluorescence experiments can be at times. That is why we have developed a set of tips to help you choose the best fluorophores for your multiplexing and multi-color experiments. Topics covered include:
10 Tips for Selecting and Using Fluorophores in IF Experiments
Interested in learning more about Immunohistochemistry? Then check out or IHC-P tips and tricks ebook.
Applies especially to emission/excitation/dichroic filters and in the case of confocal microscopes the laser lines. Getting to know your set-up ensures that you can select fluorophores that can be optimally excited and detected.
These references will help you determine the maximum excitation and emission wavelengths of the fluorophore conjugated antibodies available in your lab. Compatibility evaluation can be straight forward for certain filters, such as FITC/TRITC filters, as these have been named after the fluorophores they are meant to be used with. As members of dye families, such as Alexa Fluor® 488, have been named/numbered according to their approximate excitation maxima (in nm), these numbers also provide a rough guidance for laser and filter selection.
One defining factor of a fluorophore’s brightness is its extinction coefficient (a measurement of the probability of absorbing a photon of light); the higher the value of the extinction coefficient the brighter the fluorophore. For example the rather dim DyLight® Fluor 350 has an extinction coefficient of only 15,000 while the bright DyLight Fluor 650 dye has an extinction coefficient of 250,000 (Thermo Fisher Scientific Inc 2012).
The quantum yield is a read-out of the efficiency of the fluorescence process ( Φ is calculated by dividing the number of emitted photons by the number of absorbed photons). A 100% efficient fluorescence process would have a quantum yield of 1 (the maximum quantum yield possible). The commonly used and very bright Alexa Fluor 488 fluorophore for example has a high quantum yield of 0.92 (Molecular Probes® 2010).
Photobleaching is a photochemical destruction process that reduces the intensity of the fluorescence signal; for example FITC and R-Phycoerythrin are known to photobleach quickly. Where possible we recommend you use photostable fluorophores, such as Alexa Fluor or DyLight Fluor dyes. Alternatively, you can also reduce the intensity/exposure time to the excitation light or use mounting media containing antifade reagents. For more information about mounting media, please refer to the “Mounting coverslip section” of the Bio-Rad IHC-P tips and tricks ebook.
Many conventional fluorophores, such as FITC, are not recommended for staining protocols using acidic buffers as the fluorescence intensity signal is highly sensitive to an acidic environment (Cancer Institute, The University of Mississippi Medical Centre 2015).
Counterstains provide background contrast and put the observed staining into perspective (e.g. by visualizing nuclei). It is important that your conterstain and fluorophore staining can be easily distinguished. For example, DRAQ5™ should be used as a nuclear counterstain rather than DAPI when using antibodies conjugated to blue emitting fluorophores such as Alexa Fluor 405 or DyLight Fluor 405.
Spectral overlap is also known as bleed-through and describes the detection of one fluorophore in another fluorophore’s filter set. Bleed-through makes it difficult to observe discrete fluorescence signals and complicates the evaluation of co-localization experiments. Therefore when selecting fluorophores the emission and excitation spectra should be checked for spectral overlap with the help of a spectrum viewer. Ideally, there should be no spectral overlap between the fluorophores. Quantum dot conjugates, which have broad absorption and very narrow emission spectra, are therefore perfectly suited for multi-color experiments.
The brightest fluorophore should be reserved for the detection of the antigen with the least abundant expression level. The dimmest fluorophore should be used for detecting the most abundant antigen.
In order to verify that the observed fluorescence is a result of staining rather than an unspecific artifact we recommend you consider inlcuding these three controls:
Alexa Fluor® is a registered trademark of Molecular Probes Inc. OR, USA.
DyLight® Fluor is a trademark of Thermo Fisher Scientific Inc. and its subsidiaries.
DRAQ5™ is a trademark of Biostatus Limited.